This video demonstrates how to open the Air Tank Purge Valve on the back door of the Triton Self-Contained delivery system. It is recommended that the unit be depressurized when not in use. This vi...
A step by step guide on how to purge the suction system using KeepClear. Click here to purchase Keep Clear Cleaning Solution Copyright © ASI Medical. All rights reserved.
This video demonstrates how to remove the fan filter from the back door of your self-contained dental unit. This filter was originally introduced into the cart to buffer the sound coming from the f...
This video demonstrates how the flow control knob should sit on the needle valve stem. There should be a little gap between the stem and the knob (meaning the knob doesn't sit directly on the stem ...
This video demonstrates what to expect when unwrapping your Triton Dental System for the first time. Inside of the shipping container, you will find a red, white and blue "Customer Welcome Kit" wit...
A step-by-step guide on how to adjust the handpiece pressure on your ASI system. Copyright © ASI Medical. All rights reserved.
A step-by-step video on how to install an external amalgam separator onto your ASI system. Click Here to order an Amalgam Separator. Copyright © ASI Medical. All rights reserved.
A step-by-step instructional video on how to connect the air supply, electrical connection, optional tubing and suction, as well as Micro Evacuation connection to the Junction box from the cart umb...
A step-by-step instrucional video on how to remove/replace the suction canister in your ASI system. Copyright © ASI Medical. All rights reserved.
This video demonstrates the proper way to connect, operate and deactivate a handpiece on an ASI dental deliver unit. Please note, the demonstration is valid for all ASI units, not just the self-con...