Videos tagged 'Preventative Maintenance'
Videos that have step-by-step instructions on how to properly maintain the ASI system on a daily, weekly, and annual basis.
Disinfecting Exterior Surfaces [66-3009]
This video demonstrates the proper method of disinfecting and cleaning the exterior of the ASI dental delivery system. The cleaning procedures apply to all ASI dental delivery systems, not just the...
End of Day Maintenance: Opening Air Tank Purge Valve [66-3007]
This video demonstrates how to open the Air Tank Purge Valve on the back door of the Triton Self-Contained delivery system. It is recommended that the unit be depressurized when not in use. This vi...
End of Day Maintenance: Purging the Suction System [66-3000]
A step by step guide on how to purge the suction system using KeepClear. Click here to purchase Keep Clear Cleaning Solution Copyright © ASI Medical. All rights reserved.
End of Day Maintenance: Removing Moisture Condensation[66-3007-B]
Part of the end-of-day maintenance for the self-contained units with a compressor is to open the air tank purge valve to drain out the system of collected moisture from the day. Watch this video to...
How To Adjust Handpiece Pressure [66-3004]
A step-by-step guide on how to adjust the handpiece pressure on your ASI system. Copyright © ASI Medical. All rights reserved.
Micro and Active Irrigation- Daily/Weekly/Monthly Maintenance [66-3002]
Preventative Maintenance for Portable/Mobile, Self-Contained Dental Unit Systems that are equipped with ASI's irrigation handpieces. Copyright © ASI Medical. All rights reserved.
Oil Mist Recovery Jar Preventative Maintenance [66-3003]
The oil mist jar on your ASI cart captures excess oil from handpieces. This video shows step-by-step on how to properly maintain the oil mist jar for best performance. Click Here to order an Oil ...
Periodic Maintenance: Removing Sediment Bowl - Units manufactured 11/2019 and after [66-3005]
This video demonstrates how to remove the sediment bowl from the suction canister for maintenance. This video is intended for customers with units manufactured November 2019 and after. Copyright...
Quarterly Maintenance: Cleaning and Replacing Inline Purge Filter (Post-April 2021) [66-3011]
This video demonstrates how to remove, clean and/or replace the inline purge pump filter on the ASI Triton Dental Delivery System (for units built April 2021 and after). This maintenance should b...
Quarterly Maintenance: Cleaning or Replacing Inline Purge Filter (Pre-April 2021) [66-4036]
This video demonstrates how to remove and replace the mesh filter inside the in-line purge tubing filter bowl assembly (for units built before April 2021). The part number for the in-line filter ...
Sediment Bowl Removal - Units Manufactured before 11/2019 [66-3006]
This video demonstrates how to remove the sediment bowl from the suction canister to preform maintenance. This video is intended for customers with ASI units manufactured before November of 2019 (P...
Self-Contained Maintenance - For most units made before 2018 [66-3001]
This video demonstrates the general preventative maintenance needed for the Triton Dental Self-contained Systems. Most units manufactured prior to 2018 should follow these procedures. For units mad...